OLYMPUS CF-Q180AI colonoscope,SN:280****.
1. With image ;
2. 25% light bundle break ；
3. The insert tube is slightly aged, leakage at the hardness adjust position;
4. Biopsy tube wear;
5. Bending section deformation, down angle failure;
6. The rubber leaks.
Suggest to replace light bundle,insert tube, biopsy tube,bending section,wires, helix tube and rubber. ( No warranty for CCD)
Repair cost: $****.