OLYMPUS BF-XP160F fiber scope, SN:280****
1. Image degumming, serious color, CCD aging, a few black and gray spots;
2. 20% of the light bundle is broken;
3. Insert tube aging slightly;
4. Biopsy tube wear, not smooth;
5. Bending section, steel wire, screw tube wear;
Suggestion: replace CCD, image beam, beam, insert tube, pliers tube, snake bone, steel wire, spiral tube, rubber.
Repair cost: $****