OLYMPUS LTF-160 thoracoscope for repair,SN: 270****.
1.with image, but big risk for CCD after leakage.
2. Light bundle broken completely.
3. Insert tube with little scratch.
4. Biopsy tube abrasion and leakage.
5. Wires and helix tube abrasion.
6. Grip broken, leakage.
7. All the buttons failure.
8. Rubber leakage.
9. the whole scope leakage.
Solution: Replace CCD,light bundle, biopsy tube, wires,helix tube,grip and rubber, repair all the buttons.
Repair time: 7 working days.
Repair price: $****.
Warranty: 90 days.